What do I do if the BodyGem device does not calibrate correctly?
Remove the power cord from the side of the BodyGem unit and reinsert it. Repeat the steps in the BodyGem User Manual, making sure there are no air vents nearby. If the BodyGem device continues to display an error message, try moving to another location to perform the measurement. If for some reason you receive an Error 23 measurement, be sure the flowtube is completely flush with the BodyGem casing.